Complete Calibration Of Pyrometers
Category: PYROMETRY AND PYROMETERS
For the complete calibration
of a thermo-couple of unknown electromotive force, the new couple
may be checked against a standard instrument, placing the two bare
couples side by side in a suitable tube and taking frequent readings
over the range of temperatures desired.
If only one instrument, such as a millivoltmeter, is available,
and there is no standard couple at hand, the new couple may be
calibrated over a wide range of temperatures by the use of the following
Water, boiling point 212 deg.F.
Tin, under charcoal, freezing point 450 deg.F.
Lead, under charcoal, freezing point 621 deg.F.
Zinc, under charcoal, freezing point 786 deg.F.
Sulphur, boiling point 832 deg.F.
Aluminum, under charcoal, freezing point 1,216 deg.F.
Sodium chloride (salt), freezing point 1,474 deg.F.
Potassium sulphate, freezing point 1,958 deg.F.
A good practice is to make one pyrometer a standard; calibrate it
frequently by the melting-point-of-salt method, and each morning
check up every pyrometer in the works with the standard, making the
necessary corrections to be used for the day's work. By pursuing
this course systematically, the improved quality of the product
will much more than compensate for the extra work.
The purity of the substance affects its freezing or melting point.
The melting point of common salt is given in one widely used handbook
at 1,421 deg.F., although chemically pure sodium chloride melts at
1,474 deg.F. as shown above. A sufficient quantity for an extended
period should be secured. Test the melting point with a pyrometer
of known accuracy. Knowing this temperature it will be easy to
calibrate other pyrometers.
Next: Placing Of Pyrometers
Previous: Calibration Of Pyrometer With Common Salt